The iHealth Fingertip Pulse Oximeter offers an affordable, reliable and precise way to check your pulse rate and blood oxygen saturation levels-whenever and wherever. The Pulse Oximeter features a one-button operation to monitor your blood oxygen saturation levels (SpO2) as well as your heart rate (pulse).
Easy to Use
One button operation
Heart Rate Display
Pulse rate / heart
Calculation of arterial
Accurate and Reliable
iHealth fingertip pulse oximeter accurately
measures SpO2 (blood oxygen saturation
levels ) within ± 2% deviation.
Finger chamber is made of high quality medical silicone that is hypoallergenic and latex free.
SpO2 Levels Reach 100%
Blood Oxygen Saturation Level readings
reach up to 100%. SpO2 and Pulse Rate
readings display clearly.
Easy Battery Removal
Easy snap-off battery cover allows
easy battery removal and replacement.
iHealth Fingertip Pulse Oximeter
Measuring Oxygen Level (SpO2) and Heart Rate ( pulse)
(*Not a wireless model)
The iHealth fingertip pulse oximeter offers an affordable, reliable and precise way to check your pulse rate and blood oxygen saturation levels – whenever, and wherever. The Pulse Oximeter features a one-button operation to determine your blood oxygen saturation levels ( SpO2) as well as your heart rate (pulse).
Display Format: LCD Display
SpO2 Measuring Range: 0% ~ 100%
Pulse Rate Measuring Range: 30 bpm ~ 250 bpm
Pulse Wave Display: columniation display and the waveform display.
Power Requirement: 2x 1.5 V AAA Alkaline battery ( or using the rechargeable battery instead), adaptable range: 2.6 V – 3.6V
Power Consumption: Smaller than 30 mA
Resolution: 1% for SpO2 and 1 bpm for Pulse Rate.
Measurement Accuracy: ±2% in a stage of 70%-100% SpO2, and meaningless when the stage being smaller than 70% ±2 bpm during the pulse rate range of 30-99 bpm and ±2% during the pulse rate range of 100~250 bpm. Clinical Trial: SpO2 regression plot & Bland-Altman plot, Refer to Figure 2 & Figure 3.
Measurement Performance in Weak Filling Condition: SpO2 and pulse rate can be shown correctly when the pulse-filling ratio is 0.4%. SpO2 error is ±4%, pulse rate error is ±2 bpm during the pulse rate range of 30~99 bpm and ±2% during the pulse rate range of 100~250 bpm.
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and darkroom is less than 1%.
It’s equipped with a function switch: The product will enter the standby mode when no signal is in the product within 5 seconds.
Red light ( wavelength is 660 nm, 6.65 mW)
Infrared (wavelength is 880 nm, 6.75 mW)
*Some iHealth fingertip pulse oximeter may arrive without the iHealth logo on the package but with Andon logo. This is still an iHealth fingertip pulse oximeter but we have removed the step for printing the new package to improve production speed due to the high demand during this crisis. All other components are the same. It’s the same model.